Thin Film Aniline Point Apparatus
For aniline point and mixed aniline point determinations according to Method
B. Stirs aniline-sample mixture in a borosilcate glass thin film tube
suspended in a heating bath. Thin film of mixture flows over a light well
illuminated by a variable 6V lamp. Adjust heating rate per specifications
using accessory Powertrol Heater. When lamp filament brightens inside well,
allow mixture to cool until the two phases separate as indicated by obscuring
of the lamp filament. Consists of thin film tube; 400mL Pyrex™ beaker; cover
assembly with bath stirrer, sample pump rotor and cooling coil; 6V lamp with
line cord; and drive motor. Positive drive pulley system rotates sample and
bath stirrers. Accessory Powertrol Heater has variable stepless control and a
reference dial for repeatable control of heating rate. Porcelain refractory
top plate shields 75OW heater and has a positioning well for the Pyrex™ bath.
Low voltage receptacle in heater housing accepts line cord of 6V lamp.
Conforms to: ASTM D611; IP 2; ISO 2977; DIN 51775; FTM 791-3601
Bath Medium: 400mL of heat transfer fluid (355-000-001 mineral oil is
suitable for this application)
115V, 50/60Hz, Single Phase, 6.5A
220-240V, 50/60Hz, Single Phase, 13.4A
Thermometer Ferrules (2)
Clamps and Support Rod
Net Weight: 24 lbs (10.9kg)
Shipping Weight: 42 lbs (19.1kg)
Dimensions: 5.7 Cu. ft.